[USRP-users] LO phase alignment
john.malsbury at ettus.com
Thu Jun 21 10:33:01 EDT 2012
This feature is currently supported in the master branch. I will update
the document when it is moved to release. If you search the list
archives, you can find some information on how to use this feature.
Here's a quote from one such e-mail sent by Josh Blum:
"Use set_command_time and clear_command_time around any call you wish to
Note that this will remove the tune-to-tune offset variations. But you
will probably still need to perform some level of calibration and expect
this to change a bit with temperature, etc. I'd recommend reading the
mimo and sync app note on the kb as well.
On 06/21/2012 07:27 AM, Christophe ALEXANDRE wrote:
> Hi all,
> i want to use 2 x (N210 + SBX) for a laser telemetry application.
> i need to synchronize clocks, align sample times and also have precise
> of LO phase difference between both N210s.
> At the moment, using a mimo cable, i use the following
> method : i generate a 1.00005 GHz and inject it on both N210s
> with a center frequency equal to 1 GHz.
> I get 2 exact 50 kHz complex signals with a constant phase difference.
> Each time
> i start a measure, i get an other phase difference, still constant.
> if i change the frequency from 1 GHz to 1.1 GHz then go back
> to 1 GHz, the phase difference has changed.
> I can't use this for a telemetry application.
> i have read this application note :
> and find this :
> _Do I plan to implement beamforming or direction finding (DF)
> capability? _
> While all daughterboards can be used in a MIMO configuration, this
> does not necessarily imply there is phase alignment between the RF
> chains of one or more daughterboards. The BasicRx, BasicTX, LFRX and
> LFTX are exceptions to this, since they do not include local
> oscillators that contribute to phase ambiguity between channels.
> The Ettus Research SBX daughterboard utilizes an RF PLL that includes
> a resynchronization feature, which can be used to align the LOs across
> multiple SBXs, and multiple USRP hardware devices. Using the
> UHD (USRP Hardware Driver^(TM)), timed SPI commands can drive this
> re-synchronization feature. At the time of this writing, this feature
> is not supported in the mainline UHD. However, this feature is planned
> for release in 2012.
> I can't work without phase alignment. So my question is :
> *when could we get this feature in UHD ?*
> Christophe ALEXANDRE
> Conservatoire National des Arts et Métiers (CNAM)
> Laboratoire CEDRIC-LAETITIA
> Département EASY
> Accès 17-1-32, Case 2D2P10
> 292 rue Saint Martin
> 75141 PARIS CEDEX 03
> email : christophe.alexandre at cnam.fr
> <mailto:christophe.alexandre at cnam.fr>
> tel. 0140272699
> mob. 0651087311
> fax. 0140272994
> USRP-users mailing list
> USRP-users at lists.ettus.com
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the USRP-users